Thanks, Carl Dempsey Cadex Electronics. Most Active Software Boards: Message 5 of Let us know how it goes! When I issue the command I see the “Link” indicator on the BK appear for a few seconds and “sometimes” the unit switches into remote mode – mostly it just stays in the front-panel mode. Programming a BK precision in List Mode? It seems like this may just be the result of syntax problems.
|Date Added:||28 August 2013|
|File Size:||31.30 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 9 of I called BK several times and those guys really don’t know anything. Repeat until the battery voltage drops to a certain threshold.
Anyway, any inputs from you would be greately appreciated. I have trouble receiving kb, the byte start moves the AA and the data obtained is wrong when it comes to read it. I’m just using a simple serial port VI and am entering the 20H command which should set it to remote mode. Thanks, Carl Dempsey Cadex Electronics.
Programming a BK precision in List Mode? – Discussion Forums – National Instruments
Message 5 of Message 4 of Most Active Software Boards: When I issue the command I see the “Link” indicator on the BK appear for a few seconds and “sometimes” the unit switches into remote mode – mostly it just stays in the front-panel mode. In order to help you, what exactly are you trying to program this labvieew to do?
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Has anyone programmed a BK precision in List Mode? When I start the VI, I get the displays on the front panel showing that the input is on, the unit is in list mode, and that it’s operating in remote control.
I haven’t specifically worked with that instrument, but I still should be able to help you. I just figured out how to program the in precosion mode.
A good troubleshooting strategy would be to use NI-Spy to make sure that the syntax you’re sending out is exactly the way that you hope it too.
Message 10 of I’m trying to set the BK in List Mode to discharge a battery and different currents: If you can get that working, then using labVIEW to program it shouldn’t be much more difficult.
When I put the unit into local control I can manually check all the inputs and find that they’re all correct.
BK Precision 85XX Series Electronic Load
While doing this I learned a variety of “secrets” labvisw this dumb thing. It seems like this may just be the result of syntax problems. Message 2 of It would pabview hurt to put a return character or new line character on the trigger line. Programming a BK precision in List Mode? I’m thinking I must have some type of trigger problem. Let us know how it goes! What seems to be the issue you’re experiencing?
Hi Carlos, Could you explain a bit more about what you mean by “byte start moves the AA”?.
My VI sends all the commands in order for this unit to operate in list mode. Message 8 of If it is possible please post or send me the details carl. I find it to be quite an interesting and hopefully useful device for battery testing. I’m trying to communicate to my BK Precision Message 3 of